Albedo, also called solar reflectance, is defined as the ratio of the reflected to the global radiation. The solar albedo depends on the directional distribution of incoming radiation and on surface properties at ground level. Albedos of typical surfaces range from about 4 % for fresh asphalt, and 15 % for green grass to 90 % for fresh snow.
Using SRA20-D2 albedometer is easy. The instrument is composed of two SRA20-D2 digital secondary standard pyranometers; the upfacing one measuring global solar radiation, the downfacing one measuring reflected solar radiation. SRA20-D2 offers two types of outputs commonly used in the solar PV industry: digital via Modbus RTU over RS-485 and analogue 4-20 mA (current loop). These industry standards allow for easy data acquisition, easy read-out and error-free instrument exchange when using SRA20-D2. The working principle and specifications of the pyranometers can be found in the SR20-D2 user manual. We recommend to use SRA20-D2 in particular in applications where the highest measurement accuracy is required.
SRA20-D2 consists of two identical pyranometers model SR20-D2, one facing up, one facing down. The two sensors should be ordered with one AMF02 albedometer mounting kit. AMF02 includes a fixture with rod for mounting purposes and a glare screen. The user assembles these modular components into SRA20-D2 albedometer. Mounting hardware, tools and a mounting and fixation instruction are part of the AMF02 delivery too.
- PV monitoring with Bifacial solar modules
- High-accuracy meteorological observations
- Building physics, roof reflectance studies
- Extreme climates (tropical / polar)
Areas of Application
Meteorology / surface energy flux measurement
Solar energy / PV system performance monitoring
Longer cables, in multiples of 5 m, cable lengths above 20 m in multiples of 10 m
|Measurand||global solar radiation and reflected solar radiation|
|Measurement range||0 to 2000 W/m²|
|Optional measurand||albedo or solar reflectance ,net solar radiation|
|Included sensors||2 x identical ISO 9060 secondary standard pyranometer|
|Mounting||mounting rod with 15 x 10⁻³ m diameter|
|Calibration uncertainty||< 1.2 % (k = 2)|
|Calibration traceability||to WRR|
|Spectral range||285 to 3000 x 10⁻⁹ m|
|Rated operating temperature range||-40 to +80 °C|
|Temperature response||< ± 0.4 % (-30 to +50 °C)|
|Temperature response test and directional response test of individual instrument||Reports included|
|Standard cable length||2 x 5 m (see options)|
|Communication protocol||Modbus over 2-wire RS-485|
|Rated operating voltage||5 to 30 VDC range|
|Power consumption||< 150 x 10⁻³ W at 12 VDC (in total)|
|Output||irradiance in W/m²|
|Transmitted range of 4-20 mA output||0 to 1600 W/m²|
|Rated operating voltage of 4-20 mA output||5.5 to 40 VDC range|
|– main supply||< 150 x 10⁻³ W at 12 VDC (in total)|
|– 4-20 mA current loop||< 80 x 10⁻³ W at 12 VDC, with recommended 2 x 100 Ω shunt resistors|
|Sensors||(2 x) SR20-D2 secondary standard pyranometer|
|Included with AMF02||(1 x) glare screen; (1 x) fixture with rod; (2 x) o-ring; (1 x) conical positioner); (2 x) plug (pre-mounted); (2 x) M5x12 socket head cap screw; (1 x) M6x10 socket head cap screw; (2 x) M6x12 set screw (pre-mounted); (1 x) instruction sheet|
Frequently asked questions
A pyranometer measures the solar radiation received by a plane surface from a 180 ° field of view angle. This quantity, expressed in W/m², is called “hemispherical” solar radiation. The solar radiation spectrum extends roughly from 285 to 3000 x 10⁻⁹ m. By definition a pyranometer should cover that spectral range with a spectral selectivity that is as “flat” as possible.
In an irradiance measurement by definition the response to “beam” radiation varies with the cosine of the angle of incidence; i.e. it should have full response when the solar radiation hits the sensor perpendicularly (normal to the surface, sun at zenith, 0 ° angle of incidence), zero response when the sun is at the horizon (90 ° angle of incidence, 90 ° zenith angle), and 50 % of full response at 60 ° angle of incidence. A pyranometer should have a so-called “directional response” (older documents mention “cosine response”) that is as close as possible to the ideal cosine characteristic.
In order to attain the proper directional and spectral characteristics, a pyranometer’s main components are:
• a thermal sensor with black coating. It has a flat spectrum covering the 200 to 50000 x 10⁻⁹ m range, and has a near-perfect directional response. The coating absorbs all solar radiation and, at the moment of absorption, converts it to heat. The heat flows through the sensor to the sensor body. The thermopile sensor generates a voltage output signal that is proportional to the solar irradiance.
• a glass dome. This dome limits the spectral range from 285 to 3000 x 10⁻⁹ m (cutting off the part above 3000 x 10⁻⁹ m), while preserving the 180 ° field of view angle. Another function of the dome is that it shields the thermopile sensor from the environment (convection, rain).
• a second (inner) glass dome: For secondary standard and first class pyranometers, two domes are used, and not one single dome. This construction provides an additional “radiation shield”, resulting in a better thermal equilibrium between the sensor and inner dome, compared to using a single dome. The effect of having a second dome is a strong reduction of instrument offsets.
• a heater: in order to reduce the effect of dew deposition and frost on the outer dome surface, most advanced pyranometers have a built-in heater. The heater is coupled to the sensor body. Heating a pyranometer can generate additional irradiance offset signals, therefore it is recommended to activate the heater only during night-time. Combining a heater with external ventilation makes these heating offsets very low.
There are good reasons why pyranometers are the standard for solar radiation measurement in outdoor PV system performance monitoring.
The purpose of outdoor PV testing is to compare the available resource to system output and thus to determine efficiency. The efficiency estimate serves as an indication of overall performance and stability. It also serves as a reference for remote diagnostics and need for servicing.
The irradiance measurement for outdoor PV performance monitoring is usually carried out with pyranometers. Some standards suggest using PV reference cells. Reference cells are (with some minor exceptions) unsuitable for proof in bankability and in proof of PV system efficiency. Pyranometers are and will remain the standard for outdoor solar energy monitoring.
From a fundamental point of view:
- Pyranometers measure truly available solar irradiance (so the amount of available resource). This is the parameter you need to have for a true efficiency calculation.
- Reference cells measure only that part of solar radiation that can be used by cells of identical material and identical packaging (flat window), so the yield of a certain PV cell type. This is not a measurement that can be used in an efficiency calculation and in fact leads to several percentage points error in efficiency estimates.
The International Energy Agency (IEA) and ASTM standards for PV monitoring recommend pyranometers for outdoor PV monitoring. PV reference cells do not meet IEC 61724-1 class A requirements for irradiance measurement uncertainty: their directional response makes them systematically overestimate daily radiant exposure in J/m2 (or W·hr/m2 ) by more than 2 %, larger on hourly basis.
Choosing the right pyranometer for your application is not an easy task. We can offer assistance. But first, you should ask yourself the following questions:
- are there standards for my application?
- what level of accuracy do I need?
- what will be the instrument maintenance level?
- what are the interfacing possibilities?
When discussing with Hukseflux, our recommendation for the best suited pyranometer will be based on:
- recommended pyranometer class
- recommended maintenance level
- estimate of the measurement accuracy
- recommended calibration policy
- recommended interface
Pyranometers can be manufactured to different specifications and with different levels of verification and characterisation during production. The ISO 9060 – 1990 standard, “Solar energy – specification and classification of instruments for measuring hemispherical solar and direct solar radiation”, distinguishes between 3 classes; secondary standard (highest accuracy), first class (second highest accuracy) and second class (third highest accuracy). From second class to first class and from first class to secondary standard, the achievable accuracy improves by a factor 2.
The ISO 9060 – 1990 standard is up for revision. The new 2018 version of the standard will be slightly different from the 1990 version. The new version of ISO 9060 includes three instrument accuracy classes A, B and C, and a special extension of every class “Spectrally Flat”, which is recommended for Plane of Array (POA), albedo, and reflected solar measurements.
Our pyranometer selection guide offers practical guidelines for choosing a pyranometer. The application of pyranometers in PV system performance monitoring according to IEC 61724-1 is highlighted as an example. Sensors specific for diffuse radiation and meteorological networks are also addressed in this selection guide.
A pyranometer measures hemispherical solar radiation. When measuring in the horizontal plane this is called Global Horizontal Irradiance (GHI). When measuring in “plane of array”, next to PV panels, this is called plane of array POA irradiance.
A pyrheliometer is used to measure Direct Normal Irradiance (DNI). DNI is defined as the solar radiant flux collected by a plane unit surface normal to the axis pointing towards the centre of the sun, within an optical angular aperture. DNI is composed of the solar irradiance within the extent of the solar disk (half-angle 0.266 ° ± 1.7 %) plus some circumsolar radiation.